LIGO Document E1600114-v1
- The laser power feedthroughs at the table patch panels currently use CONNEC connectors that are not reliable due to strain induced on the backshell. Several upgrade possibilities were discussed, but the simplest option is to replace the connectors with a hard-wired screw connection inside a metal electrical box. This allows use of standard cable clamps appropriate for the large diameter cable. Patch panel will need drilled to mount box.
- 4/12/2016: approved by Dennis Coyne
Before an associated FRS (Integration issue & ECR Tracker) entry is created (in order to follow the implementation of this ECR), we need clarification - does this ECR resolve/fix the problem reported in FRS #4531:
DCC Version 3.4.1, contact