LIGO Document T1500377-v8
- In this document we describe a method for calculating frequency independent changes in the DARM actuation and sensing functions and the coupled-cavity pole frequency using one x_ctrl and one Pcal calibration line.
- Updated fig. 1 to accurately locate the "minus sign" in the loop.
Updated table two by swapping the line 1 and line 3 frequencies to reflect the error made when implemented at LLO.
Changed minus to plus in equations 5, 7, 9, and 19 to reflect the change in the placement of the minus sign in the loop diagram.
Added the beginning of Sec. 5, the for the calculation of h(t).
DCC Version 3.2.2, contact
Document Database Administrators